Unscrew the shipping protection screw of the scale. Montar el transductor. Mount the scale. Posicionamiento de los relojes comparadores. Positioning of the dial indicators.
|Genre:||Health and Food|
|Published (Last):||14 February 2019|
|PDF File Size:||6.17 Mb|
|ePub File Size:||1.70 Mb|
|Price:||Free* [*Free Regsitration Required]|
Unscrew the shipping protection screw of the scale. Montar el transductor. Mount the scale. Posicionamiento de los relojes comparadores. Positioning of the dial indicators. Alinear las caras del transductor, realizando mediciones en los extremos y en el punto central. Align the sides of the scale by indicating the ends and the center. Mounting possibilities. Remove the head securing plate red. Conectar el cable de la cabeza lectora y comprobar que la impedancia existente entre el conector y el transductor es inferior a 1 Ohmio.
Connect the reader head cable and verify that the impedance between the connector and the scale is less than 1 Ohm. It is recommended to mount some kind of protection cover as to avoid damage caused by materials or liquids falling onto the scale.
Soltar el taco 2. Reventar el agujero y roscarlo con metrica M5 3. Amarrar racor de entrada de aire 4. Remove the end-cap 2. Punch the hole through and tap it with Metric 5mm 3. Fasten the air inlet. Mount the end-cap back by adding silicone first. A partir de esta marca de referencia, una cada 50 mm en ambos sentidos. Cable Length: 3 m. From there, one every 50mm 2 inches in both directions. The outer reference mark will be located at the following distance from the ends of the measuring length.
Se debe utilizar cable de 8x0. Si Rz Ohmios: 50 mts. Longitud de cable permitida: 20 mts. A 8x0. With models other than Fagor its maximum length depends upon the line terminating resistor of the receptor unit Rz.
If Rz Ohms: 50 m. Maximum cable length: 20 m. Longitud de cable permitida: mts. Maximum cable length: m. EN Burst and fast Transients. Perturbaciones conducidas por campos en ENV Conducted disturbance induced by radio frequence fields. The information described in this manual may be subject to variations due to technical modifications. Short-link Link Embed.
Share from cover. Share from page:. More magazines by this user. Close Flag as Inappropriate. You have already flagged this document. Thank you, for helping us keep this platform clean. The editors will have a look at it as soon as possible. Delete template? Cancel Delete. Cancel Overwrite Save. Don't wait! Try Yumpu.
Start using Yumpu now! Resources Blog Product changes Videos Magazines. Integrations Wordpress Zapier Dropbox.
TRANSDUCTORES LINEALES LINEAR TRANSDUCERS
Descubra todo lo que Scribd tiene para ofrecer, incluyendo libros y audiolibros de importantes editoriales. This guide contains the basic information necessary to interface the Vantage, Maestro or ProTech control system with plant automation systems, Man-Machine-Interface MMI software, and other data logging systems using the Modbus communications interface. However, since installations may vary, these instructions may not cover all details or variations in the equipment supplied or every question that may arise during use. If a question or situation develops which is not answered directly in this guide, contact an authorized service representative for more information, or contact the factory directly for specific answers and advice. Guide Revision 1. The controller operates as the slave device, the plant system or MMI operates as the Master device.
Vantage Modbus Guide